Product technical indicators:
1.The maximum recommended flow:8 L/min
2.The flow range of nominal pressure of 7kPa: 0.5-8 L/min
3.The flow rate change under the maximum recommended flow rate with the back pressure of 7kPa: smaller 1L/min
4.Oxygen concentration when the nominal pressure of the outlet is zero(the specified concentration level is reached within 30 min after the initial start-up): The oxygen concentration is 93% ± 3% at the oxygen flow rate of 8L/min.
5.The output pressure: 30-70kPa
6.Release pressure of compressor safety valve: 250 kPa ± 50 kPa.
8.Power supply:AC230V/50Hz or AC110/60Hz or AC220V/60Hz
10.Net weight: about 29KG
12.Altitude: The oxygen concentration dose not reduce at 1828 meters above sea level, and the efficiency is less than 90% from 1828 meters to 4000 meters.
14.Minimum working time: no less than 30 minutes
15.Electrical classification: Class II equipment, Type B application part
16.Character of service: Continuous operation
17.Normal working environment: Ambient temperature range: 10℃-40℃; Relative humidity ≤80%; Atmospheric pressure range: 860hPa – 1060hPa; Note:The equipment should be placed in normal working environment for more than four hours before use when the storage temperature is lower than 5℃.
18.Temperature of oxygen outlet ≤ 46℃.
19.Recommendation: The length of the oxygen tube should not exceed 15.2 meters and cannot be folded;
20.Ingress protection rating: IPXO
21.Device type: Non-AP/APG device (cannot be used in the presence of flammable anesthetic gas mixed with air of flammable anesthetic gas mixed with oxygen or methylene).
|Place of Origin||China|
|Instrument classification||Class II|
|After-sale Service||Online technical support|
|Type||Home health care|
|Display Control||LCD Touch Screen|