High definition larger screen display，voice broadcast，original imported molecular sieve，design of portable handle，easy replacement of filter felt，design of antiskid foot pad,low noise，operation alarm
1 Disassembly of class 1 filter
Located in the back shell of the machine, the filter door cover plate is buckled down,then pulled out,the filter door cover is taken out, and the level 1 filter screen is removed. The filter screen should be cleaned according to the actual use time and environment. If there is obvious dust, it should be cleaned or replaced immediately.
2 Disassembly method of intake filter cover plate.
Located on the right side of the machine, buckle the filter door cover, pull it out and take out the filter door cover.
3 Replacement method of secondary filter felt:
After the air inlet filter cover plate is removed, the air inlet cover is rotated counter-clockwise. After the air inlet cover is loosened, the air inlet cover can be removed,and the secondary filter felt can be replaced or cleaned in time.
4 Cleaning methods:
Clean with a light detergent and rinse with clean water. It must be dry before it can be loaded into the machine.
|After-sale Service||Online technical support|
|Product name||Oxygen Concentrator|
|Keywords||Oxygen Concentrator Machine|