ZY-5ZW: 93%±3% (1L-5L oxygen flow) oxygen,professional assembly line production,low to 42dB,convenient for pregnant women, the elderly and other groups to use,CE ISO,eight-fold enhanced filtration device,high-definition screen – HD LED screen,Low noise oil-free compressor,PSA oxygen method
1 Take out the oxygen generator from the box and remove all packaging.
2 Place the machine on a flat surface with the screen facing up and use scissors to remove the tie at the bottom(this is for a fixed compressor).
3 Set up the machine after cutting the tie.
4 Remove the wearing bottle, turn off the cap counterclockwise and add cold pure water. The water level is between the “Min” and “Max” scales on the wetting bottle.
Note: The optimum installation position of humidifying bottle in oxygen generator is shown.
5 Gently tighten the cap of the wetting bottle clockwise and place the wetting bottle in the installation tank of the main oxygen generator.
6 Insert one end of the connecting pipe with the oxygen outlet of the main engine and the other end with the air inlet of the humidifying cylinder,as shown.
7 Connect the power cord:First make sure that the power switch of the main oxygen generator is off. Connect one end of the power cord to the power socket of the oxygen generator and the other end of the safe grounding socket with the output electricity.
|Place of Origin||China|
|Instrument classification||Class II|
|After-sale Service||Online technical support|
|Keywords||Oxygen Concentrator Machine|
|Power supply||220V/110V 50Hz/60Hz|
|Purity||93% Oxygen Purity|
|Oxygen flow||1L -5L/min Adjustable|